[ present & archived diaries ]
[ Home ]
- 29 March 2017
Noticed Tx attenuation is 15 dB ! It should be 4 with HPTx and 0 with LPTx.
Probably like this for some time. It says 4 dB in the header.
HV has been on now very long, since 10 March !.
- 03 April 2017
Fixed DIO interface board. A faulty latch chip DM8553 ?? was replaced. Similar to a fault on an
adjacent chip which was fixed in April 2010 by Arild. See circuit diagrams HFS134 and HFS135 for details.
- 28 May 2017
06:49 UT the receiver failed. Power supply problem
- 29 May 2017
Dynasonde stopped soundings about 0928 UT.
- 07 June 2017
Receiver power supply fixed, by temporarily replacing the NFS350 series supply (failed) with an old PC power supply.
First sounding number repeated twice.
Running with HV, BPG every 2 minutes.
[ Dynasonde page] This page is maintained by Mike,
who is solely responsible for the content, or lack thereof :-/ Last
updated 29 March-2017